Error Counter timebase source, counter timebase rate, master timebase divisor, and master timebase rate settings are inconsistent with one another. For internal counter timebase source selections, if the counter timebase rate is set, its value must match the rate corresponding to the counter timebase source. For example, 20 MHz corresponds to a rate of 20,, Hz.
- The Importance of Supply chain Security Due Diligence?
- UL PRODUCT CATEGORY;
- how to check a vin number for a atv.
- ky state police sex offender registry.
- - Document - PXI moves up the food chain.
- florida social work marriage and family.
Error Counter timebase source and counter timebase master timebase divisor settings are inconsistent with one another. Error Frequency and Initial Delay property values are inconsistent with one or more counter timebase properties. Error Initial Delay, High Time, and Low Time property values are inconsistent with one or more counter timebase properties. Error A timebase could not be selected that covers the entire range specified in the Maximum and Minimum properties. Error Two consecutive active edges of the input signal occurred without a counter timebase edge.
Use a faster counter timebase. Error Data was overwritten before it could be read by the system. Otherwise, divide the input signal before taking the measurement. Error Internal timebase could not be found that matches the rate specified in the Counter Timebase Rate property.
Set the Counter Timebase Rate property to the appropriate value for your external source. Error Pause trigger is only valid for continuous pulse generations. Change the sample mode to continuous, or do not use the pause trigger.
Ni Systems India Pvt Ltd
Error Pause trigger is only valid for event counting if sample clock is not used. Error Pause and start triggers cannot both be active in this task. Error There cannot be multiple counters in the same task for input operations. Use a separate task for each counter. Choose a frequency within this range.
- collin county texas court records search;
- Want more proof points? Check out these PXI case studies:.
- los angeles website death certificate!
- meriwether county georgia census record 1910?
- Product Details.
- August 9, 12222.
- Browse by Product Type.
Error External timebase rate must be specified to translate the delay into ticks. Set the external timebase rate, or set the delay in units of ticks.
Error Channel is not available when the module is in parallel mode. Error Your SCXI system is not set up for performing analog input operations on given channels. The SCXI module cabled to your digitizer cannot route analog signals from other modules to the digitizer, or is not configured to route them. To perform the desired operation with multiple SCXI modules and one digitizer, cable the digitizer to one of the analog input modules.
The module your channels are on is one such module.
Then, update the chassis configuration in MAX to reflect cabling change, and ensure that the cabled module is in multiplexed mode. Alternatively, you can use multiple digitizers and SCXI modules in parallel mode. For detailed information about cabling, refer to documentation. Verify that any accessories configured with this device are connected. If the problem continues, contact National Instruments Technical Support. The device might need to be recalibrated or repaired by NI.
Error Reference voltage applied for calibration is outside the range defined for calibration of this device. Ensure that the reference voltage falls within the range specified for this device. Error Reference current applied for calibration is outside the range defined for calibration of this device. Ensure that the reference current falls within the range specified for this device.
Error Reference resistance applied for calibration is outside the range defined for calibration of this device. Ensure that the reference resistance falls within the range specified for this device. Error Reference frequency applied for calibration is outside the range defined for calibration of this device. Ensure that the reference frequency falls within the range specified for this device. Error Device could not complete the calibration operation. Calibration could fail for the following reasons: 1.
The actual reference signal applied for calibration was different from the value you specified. Ensure that the reference signal applied is the same as the values that were input. The reference signal was not stable over the period of time that the hardware was being calibrated. Ensure that the reference signal specified is free of noise and does not drift over the duration of the calibration. The device is not functioning properly.
- list sex offenders aransas county texas!
- california records of birth death marriage.
- Space Details.
- GitHub - NIVeriStandAdd-Ons/Ballard-ArincCustom-Device?
- rate companies doing background checks?
Error Requested operation could not be performed because the necessary digital lines could not be reserved by SCXI. Another task might have reserved these lines previously. Error Channel could not be created. All channels must be created before the task is verified. Channels must be created before these actions can occur. Error Device to which the sensor is attached does not have an available internal excitation source.
Select another device with an available internal excitation source or supply external excitation. Error 2-wire resistance configuration is incompatible with voltage excitation. Error Completion resistance value, R1, cannot be zero if the circuit uses voltage excitation. Error Output buffer underwrite. Your application was unable to write samples to the background buffer fast enough for the device to get new samples at the specified sample rate. To avoid this error, you can do any the following: 1. Increase the size of the background buffer by configuring the buffer. Increase the number of samples you write each time you invoke a write operation.
Write samples more often. Change the data transfer mechanism from interrupts to DMA. Initially write a sufficient number of samples to satisfy the specified data transfer request condition. Reduce the number of applications that your computer is executing concurrently. In addition, if you do not need to ensure that each sample is generated once and only once, you can set the regeneration mode to allow regeneration. Error Number of channels to acquire exceeds the device maximum. Reduce the number of channels. In some cases, you can access a large number of channels if they are identically configured and created consecutively.
Refer to the documentation for more information. Error Physical channel specified does not exist on this device. Refer to the documentation for channels available on this device. Error Virtual channel cannot be created. Another virtual channel with this name already exists. The resulting value for the Sample Timebase Source Divisor is not supported by your device. Refer to the documentation for more information about these three properties. Error Hardware is not responding. Ensure your hardware is powered on and all cables are properly connected. Error Operation is not permitted while the switch device is scanning.
Error Operation is permitted only while the switch device is scanning. Error Task was created with a settling time different from the current settling time. When scanning, you must use the original settling time specified when the task was created. Error Explicit connection between the channels already exists.
Pci bus number assignment
You can make only one connection between these channels. Error Path between two switch channels is not available. Error Channel name specified is not valid for the switch device. Error Switch channels cannot be disconnected because there is no explicit path between them. Error Switch channel names cannot be duplicated in the path string. Error Leg in path cannot contain two channels that are already directly connected. Error Path contains a leg with two channels that cannot be directly connected. Error Channels used to make the connection between two endpoints must be reserved for routing.
Error Channel cannot be connected to itself. Error Connection cannot be made between the specified channels because they are connected to different source channels.